Tel.: (+49) 02207 9698-0
Fax.: (+49) 02207 4750
The supply of oxygen-air blends by so called blenders is part of the clinical standard today. These blenders are often combined with a connected flowmeter for dosing the flow.
The insufflation assembly REANIMAT-O2-Blend is a universal device for the supply of medical oxygen-air-blends both for stationary and mobile use.
Dosing of the flow is possible from 0 to 60 l/min, according to the intended clinical use.
Optionally, a suction unit can be integrated in the insufflation assembly which makes this device indispensable for various non-invasive applications of breathing gas support in pediatrics, neonatology and adult care.
Due to its modular configuration the assembly can be used in all fields where oxygenated medical blended gas is required. A combination with non-invasive respiration equipment is usually possible, provided the respiration equipment has a pressure limiting facility.
Application examples: Oxygenation of respiration units such as
• Resuscitation bag
• Emergency respirators
• CPAP devices
A second take-off supplying oxygenated blended gas is an ideal upgrade possibility.
By using a Y-piece, the sum of both flows can be made available. In this way it is possible to use one take-off for the rough selection and the second take-off for fine tuning.
Example: 0-15 l/min for rough selection and 0-6 l/min for fine tuning, or, 0-30 l/min for rough selection and 0-15 l/min for fine tuning.
Adequate devices being offered in the market usually consume 8-13 l/min of blended gas, even if the flow valves are closed and the device is obviously out of service. Such a consumption causes costs and is avoidable.
The insufflation assembly REANIMAT-O2-Blend has an on/off switch by which the gas supply can be interrupted after use. It is not necessary to put the wall plugs into park position or to close the cylinder valves. It is sufficient to simply switch off the REANIMAT-O2-Blend.
Hence the device will pay off within shortest time.
The selected parameters are of impressive consistency. If the flow is changed, the oxygen concentration remains unchanged, and if the oxygen concentration is changed, the selected flow remains constant.
The flow and concentration values remain unchanged after switching off the device. It is not necessary to close the flow valves after use – the selected values are available again when switching on the device next time.
In general it is necessary to aspirate the airway in regular intervals. Due to its modular configuration the REANIMAT-O2-Blend insufflation assembly is optionally available with an integrated suction unit.
Analogue to the insufflation assembly, the suction unit is equipped with an on/off switch, too. The selected vacuum remains constant when switching off the device. When switching on the device next time, it is not necessary to adjust the vacuum again.
This convenient combination of insufflation and suction creates a compact device in need of only one pressure gas connection each for oxygen and air. Gas sockets are not blocked unnecessarily which also contributes to a clear arrangement of the patient’s care place.
Air and oxygen from central gas pipeline
Supply from central gas and cylinder
Supply from two cylinders, air and oxygen
Automatic changeover to emergency supply possible:
In many cases it is essential to secure the gas supply for every situation.
By interconnecting the preference valves WY1475 and WY1476, that are available as accessories, the pressure gas supply through gas cylinders is automatically secured, in case the permanent compressed gas supply fails or is interrupted, for example when the patient has to be moved from one place to another place within the hospital.
|Insufflation assembly 0 - 6 l/min with blender 21 - 100 % oxygen||WY1480|
|Insufflation assembly 0 - 15 l/min with blender 21 - 100 % oxygen||WY1481|
|Insufflation assembly 0 - 30 l/min with blender 21 - 100 % oxygen||WY1482|
|2nd take-off 0 - 6 l/min for WY1480 / WY1481 / WY1482||WY1485|
|2nd take-off 0 - 15 l/min for WY1480 / WY1481 / WY1482||WY1486|
|2nd take-off 0 - 30 l/min for WY1480 / WY1481 / WY1482||WY1487|
|Suction unit 0 - 0.9 bar vacuum||WY1490|
|Oxygen preference valve, 2 inputs - 1 take-off||WY1475|
|Air preference valve, 2 inputs - 1 take-off||WY1476|
|Y-piece for inter-connecting two blended gas take-offs||WY1478|
|Connection hoses to be ordered in addition:|
|Oxygen pressure hose, white, ISO32, NIST fitting, 1.5 m||WY1405|
|Oxygen pressure hose, white, ISO32, NIST fitting, 3 m||WY1411|
|Oxygen pressure hose, neutral black, ISO32, NIST fitting, 1.5 m||WY1406|
|Oxygen pressure hose, neutral black, ISO32, NIST fitting, 3 m||WY1412|
|Air pressure hose, black-white, ISO32, NIST fitting, 1.5 m||WY1407|
|Air pressure hose, black-white, ISO32, NIST fitting, 3 m||WY1438|
|Air pressure hose, neutral black, ISO32, NIST fitting, 1.5 m||WY1408|
|Air pressure hose, neutral black, ISO32, NIST fitting, 3 m||WY1439|
|Country-specific connection hoses available on request.|
|Pressure gas supply||Medical oxygen, 3.4 to 5 bar (50 - 73 psi)Pure medical compressed air, 3.4 to 5 bar (50 - 73 psi)|
|Connection thread for pressure gas||Gas-specifically NIST according to DIN EN 739|
|Blender||21% to 100% ± 3% Fi=2 accuracy|
|Gas failure alarm||1.5 to 2 bar between the inlet pressures|
|Gas consumption oxygen and air||1 to 8 l/min. plus selected take-off|
|Gas consumption for the suction unit||0 to 30 l/min|
|Vacuum of suction unit||0 to -0.9 bar vacuum, infinitely variable|
|Blended gas take-off base device||WY1480: 0 - 6 l/minWY1481: 0 - 15 l/minWY1482: 0 - 30 l/min|
|Second take of blended gas (option)||WY1485: 0 - 6 l/minWY1486: 0 - 15 l/minWY1487: 0 - 30 l/min|
|Classification according Directive 93/42/EWG||IIa|
ERROR: Content Element with uid "1737" and type "menu" has no rendering definition!