For treatment of the neonatal hyperbilirubinemia the phototherapy device BILI-COMPACT delivers the high-effective "Blue-BAM spectrum by 10 energy saving bulbs. By the high utilization of intensity the treatment times are shortened considerably
The device operates in complete silence and practically no heat is emitted in the direction of the patient. Therefore it can just be placed on the canopy of warming bed THERMOCARE K, order No. WY2030 or of incubators THERMOCARE Vita order No. WY2401 and WY2402. It is also possible to place it on incubators of other brands with suitable canopy (up to 5° slant, load approx. 5 kg), without influencing the temperature control
For infant care systems with radiant warmer and without canopy the phototherapy device BILI-COMPACT is used with a height-adjustable stand which is mobile on four antistatic castors, two of them can be kick-stopped. The flexible swivel head allows to precisely direct the Blue-BAM radiation to the patient. So it can be brought very close to the patient and the usual radiation stray is minimized.
Phototherapy device with 10 energy saving lamps blue BAM/PL9/52, wave length spectrum 460 nm, integrated hour counter, rubber feet and handles.
Order No. WY1816
Bottom side of the device height-adjustable to 1315 to 1675 mm
Order No. WY2003
Rotation of device 360°
Vertical adjustment ± 45°
Depth 525 mm Width 220 mm Height 90 mm Weight 4,8 kg Minimum distance to the patient pad 200 mm Equipment of lamps (energy-saving fluorescent lamps) 10 pieces
Operating voltage / power supply ~230 V / 50 Hz Max. power input 140 W / 0,6 A Wave length spectrum 460 nm 350 mm 16 W/m2 250 mm 20 W/m2 200 mm 30 W/m2 350 mm 400 x 700 mm 200 mm 300 x 600 mm
|Phototherapy device||White RAL 9010|
|Base of mobile stand||White RAL 9002|
Depth 560 mm Width 560 mm Width 1535 to 1935 mm Weight 16 kg Necessary underpin height > 95 mm Castors Ø 52 mm / kickstop 4 / 2 Bottom side of device height-adjustable 1315 to 1675 mm
Protection class 1 Device type B MDD class IIa EN 60601-1:2006 Yes EN 60601-2-50:2010* Yes * Particular requirements for the basic safety of infant phototherapy equipment
|Preventive maintenance (recommended)||Yearly|